Towards unblinding the surgeons: Complex electrical impedance for electrode array insertion guidance in cochlear implantation
The complications during electrode array insertion in scala tympani for cochlear implantation may cause trauma, residual hearing loss and affect speech outcomes. The inner ear is like a black box for surgeons during the insertion process with no real-time feedback and must rely on radiation- based extraoperative imaging. Impedance measurement of electrodes during insertion is a simple yet effective method to assess array position. For this, an impedance meter has been designed which can measure magnitude (|Z|), phase (θ), real (R) and imaginary (Xc) parts of impedance. A switching circuit can sequentially scan all electrode pairs at regular intervals during insertion. An Evo straight electrode array is inserted in a transparent 2:1 scaled up 2D cochlear model (11 trials) filled with 0.9% saline using a 3-degrees-of- freedom actuation system. Bipolar impedance measurements of 8 pairs (40 samples each) are taken at regular intervals during 25 mm insertion at speed of 0.05mm/sec. A notable increase in |Z| and R is observed in the apical 3 electrode pairs when they first get in to contact with the lateral wall. At the same time, the phase gets less negative (more resistive impedance) and Xc increases (less capacitance). These results show that impedance can be used for electrode array localization in cochlea and impedance change due to electrode proximity to different materials can have application in other electrode implants.
Clark, J.R., Leon, L., Warren, F.M., and Abbott, J.J. (2012). “Magnetic guidance of cochlear implants: Proof-of-concept and initial feasibility study,” J. Med. Devices, 6(3). doi: 10.1115/1.4007099.
Dhanasingh, A. and Jolly, C. (2017). “An overview of cochlear implant electrode array designs,” Hear. Res., 356, 93–103. doi:10.1016/j.heares.2017.10.005.
Eshraghi, A.A., Nazarian, R., Telischi, F.F., Rajguru, S.M., Truy, E., and Gupta, C. (2012). “The cochlear implant: Historical aspects and future prospects,” Anat. Rec., 295, 1967–1980. doi: 10.1002/ar.22580.
Giardina, C.K., Krause, E.S., Koka, K., and Fitzpatrick, D.C. (2018). “Impedance measures during in vitro cochlear implantation predict array positioning,” IEEE. Trans. Biomed. Eng., 65, 327–335.
Kontorinis, G., Lenarz, T., Sto ̈ver, T., and Paasche, G. (2011). “Impact of the insertion speed of cochlear implant electrodes on the insertion forces,” Otol. Neurotol., 32, 565–570. doi: 10.1097/mao.0b013e318219f6ac.
Pile, J., Sweeney, A.D., Kumar, S., Simaan, N., and Wanna, G.B. (2017). “Detection of modiolar proximity through bipolar impedance measurements,” Laryngoscope, 127, 1413–1419. doi: 10.1002/lary.26183.
Rau, T.S., Hussong, A., Leinung, M., Lenarz, T., and Majdani, O. (2010). “Automated insertion of preformed cochlear implant electrodes: Evaluation of curling behaviour and insertion forces on an artificial cochlear model,” Int. J. Comput. Assist. Radiol. Surg., 5, 173–181.
Rebscher, S.J., Hetherington, A., Bonham, B., Wardrop, P., Whinney, D., and Leake, P.A. (2008). “Considerations for the design of future cochlear implant electrode arrays: Electrode array stiffness, size and depth of insertion,” J. Rehabil. Res. Dev., 45, 731.
Richard, C., Fayad, J.N., Doherty, J., and Linthicum Jr, F.H. (2012). “Round window versus cochleostomy technique in cochlear implantation: Histological findings,” Otol. Neurotol., 33, 1181. doi: 10.1097/mao.0b013e318263d56d.
Roland, P.S. and Wright, C.G. (2006). “Surgical aspects of cochlear implantation: Mechanisms of insertional trauma,” In Cochlear and Brainstem Implants (Karger Publishers), 64, 11–30. doi: 10.1159/000094642.
Tan, C.T., Svirsky, M., Anwar, A., Kumar, S., Caessens, B., Carter, P., Treaba, C., and Roland Jr, J.T. (2013). “Real-time measurement of electrode impedance during intracochlear electrode insertion,” Laryngoscope, 123, 1028–1032.
Zhang, J., Bhattacharyya, S., and Simaan, N. (2009). “Model and parameter identification of friction during robotic insertion of cochlear-implant electrode arrays,” Robot. Autom., 2009, 3859–3864. doi: 10.1109/robot.2009.5152738.
Zhang, J., Wei, W., Ding, J., Roland Jr, J.T., Manolidis, S., and Simaan, N. (2010). “Inroads toward robot-assisted cochlear implant surgery using steerable electrode arrays,” Otol. Neurotol., 31, 1199–1206.
How to Cite
Authors who publish with this journal agree to the following terms:
a. Authors retain copyright* and grant the journal right of first publication with the work simultaneously licensed under a Creative Commons Attribution License that allows others to share the work with an acknowledgement of the work's authorship and initial publication in this journal.
b. Authors are able to enter into separate, additional contractual arrangements for the non-exclusive distribution of the journal's published version of the work (e.g., post it to an institutional repository or publish it in a book), with an acknowledgement of its initial publication in this journal.
c. Authors are permitted and encouraged to post their work online (e.g., in institutional repositories or on their website) prior to and during the submission process, as it can lead to productive exchanges, as well as earlier and greater citation of published work (See The Effect of Open Access).
*From the 2017 issue onward. The Danavox Jubilee Foundation owns the copyright of all articles published in the 1969-2015 issues. However, authors are still allowed to share the work with an acknowledgement of the work's authorship and initial publication in this journal.